Operating System – Windows 7, Windows 8, Windows 8. If feasible, substitute a small disc type capacitor of approximately 50 picofarsds for the removable tip on the probe. This product was sold by a reseller. The item may be a factory second or a new, unused item with defects. The Combining Cases are full-module unita which accept various combinations of submodular units. The input signal is attenuated to produce a maximum of about 15 millivolts at the modulator input. Provides DC voltage output proportional to meter indication for driving external recorder.
|Date Added:||17 November 2011|
|File Size:||19.29 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
We retain the right to refuse any refund if we feel the 401c policy is being abused. The additional equipment will also be required: Model C should read between 0.
The resistors that are placed in the circuit by the RANGE switch program the amplifier gain to compensate for the non-linear response of the AC probe. This control provides adjustment for zero-setting the meter before making AC voltage measurements.
To avoid surface contamination of the printed circuit, clean with weak solution of warm water and mild detergent after repair. This product was designated to be returned to HP. A close-fitting metal shield or bushing should be arranged to ground the outer cylinder of the probe to the outer conductor of the transmission line. Should it be necessary to use the Model AR Voltmeter Calibrator to conduct these Performance Checks, the arrangement described in Figure will provide the necessary voltage values required.
For all measurements at higher frequencies, hold the molded nose of the probe as far from the external ground piane or from object at ground potential as can conveniently be done.
HP c Voltmeter With a AC Probe hz / Volts | eBay
Equipment having similar characteristics may be substituted for items listed. All items are packed professionally with bubble wrap and sturdy boxes. This two-position slide switch sets the instrument to accept either or volt AC primary power. Adjust dc power supply for System voltmeter readings listed in Table The probe contact with tip removed can then contact the other terminal of the capacitor for the measurement.
HP Agilent 410C Multi-mode Voltmeter With Probes
Place spacer clamps on the two end instruments see step 4, Figure and push the combination into the frame. Chopper frequency will vary with f.
Place the spacer clamps between instruments as shown in step 3, Figure Wrap instrument in heavy paper or before placing in an inner container. Conduct a visual check of the Ho C for possible burned or loose components, loose connecmight tions, or any other obvious conditions which suggest a source of trouble.
DC voltage, DC jp and resistance knobs and indicators are in black; AC voltage controls and indicators are in red.
If a component is obviously damaged, clip leads as close to component as possible and then remove. At frequencies above megacycles the distance between the point of voltage measurement and anode of the probe diode must be made as short as possible. This type of connection is likely to 410cc some increase in the 4110c wave ratio of the line at higher frequencies.
To check Model C frequency response from 10 Mc to Mc, use arrangement described in Figure Figure describes the function of all front and rear panel controls, connectors and indicators.
When required, check power supply voltages as outlined in Paragraph You are responsible for return shipping costs. Open download list The Part Number for the assembly is silk screened on the interior of the 4100c board to identify it. Let HP identify any out-of-date or missing drivers and software.
HP Agilent C Multi-mode Voltmeter With Probes | eBay
Remove selected files Download files. If meter pointer moves to the left during this action, repeat 410v c and e.